- Open Access
Challenges in paper-based fluorogenic optical sensing with smartphones
© The Author(s) 2018
- Received: 19 February 2018
- Accepted: 27 April 2018
- Published: 4 May 2018
Application of optically superior, tunable fluorescent nanotechnologies have long been demonstrated throughout many chemical and biological sensing applications. Combined with microfluidics technologies, i.e. on lab-on-a-chip platforms, such fluorescent nanotechnologies have often enabled extreme sensitivity, sometimes down to single molecule level. Within recent years there has been a peak interest in translating fluorescent nanotechnology onto paper-based platforms for chemical and biological sensing, as a simple, low-cost, disposable alternative to conventional silicone-based microfluidic substrates. On the other hand, smartphone integration as an optical detection system as well as user interface and data processing component has been widely attempted, serving as a gateway to on-board quantitative processing, enhanced mobility, and interconnectivity with informational networks. Smartphone sensing can be integrated to these paper-based fluorogenic assays towards demonstrating extreme sensitivity as well as ease-of-use and low-cost. However, with these emerging technologies there are always technical limitations that must be addressed; for example, paper’s autofluorescence that perturbs fluorogenic sensing; smartphone flash’s limitations in fluorescent excitation; smartphone camera’s limitations in detecting narrow-band fluorescent emission, etc. In this review, physical optical setups, digital enhancement algorithms, and various fluorescent measurement techniques are discussed and pinpointed as areas of opportunities to further improve paper-based fluorogenic optical sensing with smartphones.
- Paper microfluidics
- Point-of-care diagnostics
- Fluorescent nanotechnology
- Smartphone integration
1.1 Point-of-care diagnostics on silicone-based substrates
Point-of-care (POC) diagnostics are rapid, low-cost, mobile tests that can be conducted in resource-limited environments by little to non-trained personnel. Conducting tests on site allows for faster response time, which in turn increases better opportunities for proper and adequate treatment [1, 2]. The incorporation of microfluidics with POC tests add complexity and versatility to the assays due to controlled flow in discrete spaces, reduction in sample volume, minimized handling of reagents, and ability to run parallel comparison analysis .
Silicone-based polymers, specifically polydimethylsiloxane (PDMS), are widely used in the fabrication of microfluidic POC platforms. PDMS provides an optically transparent (230–1100 nm), flexible, nontoxic, and low-cost material. However, when untreated, PDMS surface is a relatively hydrophobic material, which can be problematic in controlling flow especially under low pressure conditions. Also, irreversible protein adsorption to PDMS surface can eventually lead to a nonfunctional device. Although many surface modification techniques have been developed to make PDMS surface hydrophilic, induced hydrophilic states are only temporary: PDMS’ inherent hydrophobicity will return after a period of time [4, 5]. Furthermore, when developing unique infrastructures such as on-chip pumps and valves, PDMS lithography fabrication techniques require a clean room access, which in turn can become complex and expensive.
1.2 POC diagnostics on paper substrates
Alternatively, paper can be used as a microfluidic platform. It is inexpensive, easy to chemically modify (i.e. nitrogen functionalized cellulose is commonly used for biological samples), easy to fabricate, store, and transport . Inkjet processing is also a well-studied and commercially available method for easy functionalization of paper substrates. A widely known and highly used paper-based lateral flow assay (LFA) example is the colorimetric pregnancy test strip. Sandwich immunoassays are the most popularly utilized in paper-based LFAs. Immobilized antibodies on a paper substrate binds with target antigens if present in the specimen. Following, secondary antibodies conjugated to gold nanoparticles act as a reporter and bind to the immobilized antibodies on paper with captured antigens. The resulting positive reaction causes an aggregation-induced pink color appearance . This coloration is due to a spatially dependent optical property of gold nanoparticles, known as the surface plasmon resonance band .
However, colorimetric LFAs are binary, i.e. yes-or-no assays, thus difficult to quantify in a reproducible manner. In addition, they show little potential for multiplexing capabilities since they can conduct only one assay at a time. Also, commercially available LFAs require high concentration of target of interest in order to obtain a reliant signal that may not be within the normal or low physiologically relevant levels . As an example, commercially available LFAs for the detection of thyroid stimulating hormone (TSH) has a limit of detection of > 5 mIU/L, which fails to detect the normal and low concentrations of TSH (i.e. hyperthyroidism) in human blood serum . Similarly, commercially available nitrite LFAs for recognizing Escherichia coli from urine (for detecting urinary tract infection), as well as Neisseria gonorrhoeae, the most common cause of sexually transmitted disease (STD) infection of the urogenital tract, has shown a limit of detection of 106 CFU/mL . Bacteria concentration among the urinary tract infection patients can be as low as 102–103 CFU/mL in adults  and even less in children . Also in a thorough analysis of commercial assays for detection of Cryptosporidium in fecal samples, the ImmunoCard STAT! LFA platform failed to detect all 12 samples with < 175 organisms per 10 μL sample and had problems with interpretation due to low band intensity .
1.3 Fluorescent nanotechnologies lower limit of detection for POC diagnostics
Nanotechnologies combined with fluorescence detection has demonstrated the lowering of limit of detection down to the single cell or picogram protein resolution [14–17]. Implementation of fluorescent nanotechnologies such as quantum dots [18–20] and nanoclusters [21–23] have several advantages over more traditional colorimetric sensors. Nanoscale sensors can easily be tuned to respond to specific excitation wavelengths, by varying shape, size, and length. Nanostructures possess large surface areas for accommodating increased amount of bioreceptor immobilization, and this in turn results in increased sensitivity and much lower limit of detection. Nanoparticles are also highly stable and do not photobleach as easily in comparison to traditional fluorescent dyes [24, 25]. The resulting emission spectrum from these nanoparticles results in increased sensitivity, which in turn shows decreased signal-to-noise ratios. Lastly, materials at which these nanotechnologies are comprised of, such as carbon [26, 27] and gold [28, 29] show superior biocompatibility in complex biological matrices. Fluorogenics in combination with paper-based microfluidic devices, reduces cost, simplifies manufacturability, and improves ease of disposability . Therefore, fluorescent nanotechnologies on paper-based platforms have become an extremely attractive option in biological and chemical sensing .
1.4 Smartphone integration into paper-based fluorogenic optical sensing
Evidently, the next step is to integrate these fluorescent nanotechnologies on paper-based platforms in conjunction with smartphone optical sensing, utilizing its flash as a light source, its cameras as an optical detector, and potentially its software application for data processing. As a result, an easy-to-use, point-of-care, yet extremely sensitive handheld platform can be developed. Smartphone integration has shown numerous advantages and opportunities in its use as a detector and user interface platform in POC assays. Smartphones allows for network connection and access, on-board processing, and application in resource-limited settings [30–34].
The number of publications demonstrating fluorescent nanotechnology with smartphone sensing also showed a similar trend—slow increase followed by a decrease in 2017 (Fig. 1; yellow line), suggesting complications and challenges in demonstrating fluorescent nanotechnology with smartphone sensing. The number of publications demonstrating all of the above—smartphone sensing of fluorescent nanotechnology on paper-based platform is significantly smaller than overall smartphone sensing (Fig. 1; light green line), again demonstrating its complications, although this number has continued to increase over time.
Although there have been many publications and reviews attesting to these outlined advantages for paper-based fluorescent platforms, there has been a scarce discussion on the challenges, technical limitations, and alternative methods of its fluorescent optical detection, especially in conjunction with smartphone sensing. This paper will review and discuss the problems that are involved when combining fluorescent nanotechnologies, paper-based platforms, and smartphone sensing, as well as the methods that are being developed and practiced to address and overcome them.
2.1 Digital enhancement
Currently, complementary metal oxide semiconductor (CMOS) array is the most widespread image sensor for smartphones. It utilizes Bayer color filter arrays (CFA) that collect red, green, and blue (RGB) values from a grid-like structure. The raw data pixel values are processed through a demosaicing algorithm, which fill in missing RGB values through interpolation techniques, resulting in an RGB image. Following a demosaic algorithm, a denoising algorithm is then applied [35, 36]. Data image processing widely varies between smartphone models and brands. Along with data processing, physical properties between CMOS image sensors also vary by models and brands. Recently, Fontaine  released a well-organized and detailed publication outlining the different CMOS schematics among smartphone models and brands as well as the evolution of the technology over the years. Such variances include the spatial resolution between metal aperture walls, color filters used, and optical stacking thickness.
In most paper-based point-of-care assays that utilize fluorescence, pixel intensities are extracted and converted into a concentration of a target of interest, i.e., molecule [38–40], protein [23, 41], whole organism , or nucleic acid [43, 44]. When trying to control and maintain constant lighting, on-board default camera settings on smartphones can be problematic, as they are constantly trying to adjust white balance, focus and exposure during multiple assays. An especially concerning problem is the camera’s ability to resolve very small points of interests such as the test line on a LFA or fluorophore-loaded areas on paper microfluidic devices .
To further refine fluorescent images captured, it is also common to apply digital filters. With use of digital filters there is no physical adjustments required to the overall optical setup. This in turn offers a low cost and simple method for correcting undesirable flaws. In fluorescence imaging, excessive crosstalk, which is the inconsistent recognition from left to right views resulting in a blurry effect, is a typical correction to address [45, 46].
There has also been numerous publications investigating better extraction and interpretation of measured RGB pixel values from the smartphone captured images on paper-based platforms. Different color spaces have been deployed to enhance paper-based pixel intensities . In Shen et al.'s color conversion analysis and quantification of colorimetric pH test strips , a more sensitive and accurate method was developed utilizing a 12 region reference chart to account for variability in lighting conditions. The group also alluded to its use in fluorescence paper microfluidic data, although it has not yet been demonstrated. Yetisen et al.'s image processing algorithm of transforming RGB values into non-linear, linear, tristimulus, then into 2D CIE 1921 chromaticity space  showed improvements in mitigating variability due to focus, angle, lighting, and sensor type. The applied algorithm was also demonstrated using two different model phones, iPhone 5 (8 MP camera) and Samsung I5500 Galaxy 5 (2 MP camera), therefore proving interphone adaptability. McCracken et al.  demonstrated the use of a triple-reference point normalization as well as fast-Fourier transform pre-processing using two different smartphone models (iPhone 5S and Nexus 5X). The developed image process reduced spatial variability due to inconsistent paper surface, shadows, and uneven background reflectance for paper-based microfluidic assays using absorbance, quenching, and scattering measurements.
2.2 Ratiometric FRET
Fluorescence resonance energy transfer (FRET) is the mechanism in which a fluorescent signal is produced due to a transfer of electrons from a donor fluorophore to an acceptor fluorophore that is within Angstrom proximal distances. As a result, two distinctive wavelength peaks are generated and can be measured in a ratiometric manner. In order for this to be efficiently monitored, peak excitation and emission wavelengths must be sufficiently separated, while having an overlap in donor emission and acceptor excitation spectra [49–51].
Ratiometric measurement is an attractive property to measure due to its inherent ability to correct for environmental factors (such as varied lighting conditions and/or optical transparency of medium; especially useful on paper-based platforms) and to self-calibrate [52, 53]. Fluorescent dyes, although have been popularly used to demonstrate FRET-based ratiometric assays, can be easily susceptible to photobleaching . More recently, the use of quantum dots [53, 55, 56] and gold nanoclusters  have been more favorable choices for FRET-based sensors due to its photostability and superior intensity from background and undesired autofluorescence .
3.1 Light sources
Paper-based microfluidics has proven the ability to provide flow without the use of external pumps or high voltage–power (necessary in most silicone or PDMS-based microfluidic devices), but rather by spontaneous capillary action amongst paper fibers (also known as wicking). With the addition of fluorescent detection method for an analyte (protein, cell, or nucleic acid), a light source is required. When using fluorogenics, specific excitation wavelengths are important in order to obtain the desired emission spectra. This being the case, most smartphone-based fluorescent assays incorporate external light sources, the most popular being a handheld UV lamp or separate LED [52, 59, 60]. Taking this into account, various platforms have been designed to accessorize and power an external light source with a specific wavelength needed for fluorescent sensing on paper platforms.
3D-printed plastic attachments that are custom-fitted to a smartphone are widely used, where a built-in LED can be housed along with additional reflectors, collimators, and filters to improve signals [61–63]. Such attached enclosure provides a controlled environment in terms of lighting and spatial distances to improve reproducibility between assays. However, smartphone dimensionality and availability vary greatly between manufacturers and models (also by the use of protective cases and covers), making custom attachments undesirable due to its poor adaptability. Along with using an external light source, an external power source is also required.
Instead of using an external light source, the white on-board LED flash on a smartphone can be also used as an excitation source to create a fully smartphone-integrated platform [67–70]. However, band-pass or low-pass fitted filters are often used to separate out exclusive wavelengths for excitation, as smartphone flashes generate “white” light [69, 71].
3.2 Optical filters
Typical means of isolating fluorescence emission include the use of optical filters. Using a low-, high-, band-pass, or other filters inserted prior to the receiving detector (i.e. CMOS array sensor, the most common camera used for smartphones) increases selectivity of emitted fluorescent light. Not only can these filters differentiate wavelengths, but can also serve as a mechanical method for controlling unwanted scattering and diffracting light. Two types of optical filters are commonly used for fluorescence detection. The first type is an absorption filter, in which absorption at the excitation wavelength is desirable and absorption at the emission wavelength is undesirable. In contrast, an interference filter has low absorption at the excitation wavelength and high absorption at the emission wavelength.
In comparison, absorption filters are comprised of one single layer. Absorptivity can be adjusted by the overall thickness of the filter and can be modeled using the Beer–Lambert law, A = ε × l × c, where ε is the characteristic molar absorptivity of the filter, l is the path length or thickness of the filter, and c is the concentration of the absorbing material [72, 74]. Absorption properties can also be controlled by the addition of dyes such as Sudan II [75, 76], Orasol Red BL, KMPR® 1005 epoxy-based photoresist , Aptina green1, and Aptina red1 .
Other unique filters can also be used to further enhance fluorescent images collected. Lee et al.  demonstrated the use of a silo-filter comprised of metal lattices, which were used as dividends for individual pixels and light guides for fluorescent light to penetrate an absorptive, thick filter layer. The silo-filter’s metal surfaces contributes an enhanced scattering and reflectance effect, improving transmittance and overall background rejection. Photonic structures is also another widely used filtering component for controlling fluorescence emission by means of specifically patterned surfaces on gold  and plastic . In an optofluidic chip developed by Ricciardi et al. , a fluorescence immunoassay was demonstrated for the detection of actin-actin antibody complexes with superior repeatability and limit of detection. They utilized unique photonic structures for controlled light radiation into a fluorescence microscope apparatus. Similarly, Schudel et al.  developed a silicone-based microfluidic chip array that utilized actuate-to-open valve mixing and photonic crystal nanostructures to detect the binding of IgG to various proteins in an immunofluorescent assay using a charge-coupled device (CCD) array sensor.
It is notoriously known that cellulose-based paper substrates exhibit autofluorescence. Cellulose paper is strongly excited with UV, followed by blue, generating blue to green emissions . Therefore an unwanted background autofluorescence, along with paper’s reflection (back scattering), must be addressed. Also with the use of biological samples, autofluorescence and back scattering light from paper surfaces can be even more problematic [84, 85].
4.1 Pulse excitation and time-resolved detection
4.2 Autofluorescence indexing
Up until now, what has been discussed were methods practiced to avoid, normalize, or subtract the inherent autofluorescence of paper matrices. In a recent publication done by Shah and Yager , a systematic “autofluorescence index” was proposed using excitation–emission matrices for screening and selecting paper substrates for low autofluorescence when developing assays. Conventionally, primarily and solely spectral overlap is considered between target-induced fluorescence and paper over fluorescence. In Shah and Yager study, 12 different paper matrices, including nitrocellulose, glass fiber, and cellulose that are commonly used, were measured and evaluated using developed autofluorescence index equations. The proposed quantification of autofluorescence was further investigated and demonstrated using a quantum dot lateral flow immunoassay for detection of influenza A nucleoprotein. It was concluded that paper matrices with lower calculated autofluorescence indices had lower limits of detections.
4.3 Use of NIR
Autofluorescence of paper can also be avoided by using longer wavelengths, e.g. near infrared (NIR) or infrared (IR) [90–93]. Yu and White  observed that background autofluorescence of paper was reduced using 785 nm excitation in assaying Rhodamine 6G, organophosphate malathion, heroin, and cocaine from a surface-enhanced Raman spectroscopy dipstick swab. Similarly, Ju et al.  found using longer IR or NIR wavelengths as the excitation source reduced autofluorescence as well as undesired back scattering. Their paper-based platform utilizing lanthanide-doped LiYF4 upconversion nanoparticles demonstrated a limit of detection of 3.6 fmol of DNA. Doughan et al.  used a 980 nm NIR excitation to reduce background noise that usually corresponds with UV or visible wavelength excitation.
However, its implementation in paper-based assays is uncertain due to the spatial limitation between sample and CMOS sensor—how can we maintain precise distance between paper and CMOS sensor? In addition, the platform must be transparent or have efficient transmittance, while paper is obviously not. Overall, the CMOS imager technology shows promising wide field of view, lens-free translation for automation, and downsizing of traditional fluorescent optical setups. In addition, automatic correction, focus, and white balance variability that is normally associated with smartphones can be finely controlled or completely bypassed with the use of contact CMOS technology [79, 84]. However, such demonstration on paper platforms is yet to be seen, leaving smartphone sensing as a better choice for the time being.
Overall, fluorogenic-based detection, especially in conjunction with nanotechnology, has shown increasing trends in translation to low-cost and easy-to-use paper-based platforms due to its superior emission intensities, assay specificity, time-dependent resolvement, and superior storage life. Smartphone-based optical sensing of fluorescence emission is of recent development and has shown numerous technical and physical limitations, especially on paper platforms. However, recent implementation of filters, advanced image processing, and unique platform development show improved results to better address such challenges. Hopefully in the near future, the benefits of contact CMOS imagers could also be incorporated into smartphone-based fluorescence sensing on paper-based platforms.
JYY conceived the concept with input from THU. THU surveyed and analyzed the literature with input from JYY. THU wrote the first draft of manuscript with revisions from JYY. Both authors read and approved the final manuscript.
The authors declare that they have no competing interests.
Consent for publication
Ethics approval and consent to participate
This work was supported by Cardiovascular Biomedical Engineering Training Grant from US National Institutes of Health (NIH), Grant Number T32HL007955.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- G.A. Posthuma-Trumpie, J. Korf, A. van Amerongen, Lateral flow (immuno)assay: its strengths, weaknesses, opportunities and threats. A literature survey. Anal. Bioanal. Chem. 393(2), 569–582 (2009). https://doi.org/10.1007/s00216-008-2287-2 View ArticleGoogle Scholar
- P. Yager, T. Edwards, E. Fu, K. Helton, K. Nelson, M.R. Tam, B.H. Weigl, Microfluidic diagnostic technologies for global public health. Nature 442, 412–418 (2006). https://doi.org/10.1038/nature05064 View ArticleGoogle Scholar
- S.K. Sia, L.J. Kricka, Microfluidics and point-of-care testing. Lab Chip 8(12), 1982–1983 (2008). https://doi.org/10.1039/B817915H View ArticleGoogle Scholar
- S. Kumar, S. Kumar, M.A. Ali, P. Anand, V.V. Agrawal, R. John, S. Maji, B.D. Malhotra, Microfluidic-integrated biosensors: prospects for point-of-care diagnostics. Biotechnol. J. 8(11), 1267–1279 (2013). https://doi.org/10.1002/biot.201200386 View ArticleGoogle Scholar
- J. Zhou, D.A. Khodakov, A.V. Ellis, N.H. Voelcker, Surface modification for PDMS-based microfluidic devices. Electrophoresis 33, 89–104 (2011). https://doi.org/10.1002/elps.201100482 View ArticleGoogle Scholar
- K. Scida, B. Li, A.D. Ellington, R.M. Crooks, DNA detection using origami paper analytical devices. Anal. Chem. 85(2), 9713–9720 (2013). https://doi.org/10.1021/ac402118a View ArticleGoogle Scholar
- W. Zhao, A. van den Berg, Lab on paper. Lab Chip 8(12), 1988–1991 (2008). https://doi.org/10.1039/b814043j View ArticleGoogle Scholar
- J.Y. Yoon, Introduction to biosensors: from electric circuits to immunosensors, 2nd edn. (Springer, New York, 2016). https://doi.org/10.1007/978-3-319-27413-3 View ArticleGoogle Scholar
- D.J. You, T.S. Park, J.Y. Yoon, Cell-phone-based measurement of TSH using Mie scatter optimized lateral flow assays. Biosens. Bioelectron. 40, 180–185 (2013). https://doi.org/10.1016/j.bios.2012.07.014 View ArticleGoogle Scholar
- S. Cho, T.S. Park, T.G. Nahapetian, J.Y. Yoon, Smartphone-based, sensitive µPAD detection of urinary tract infection and gonorrhea. Biosens. Bioelectron. 74, 601–611 (2015). https://doi.org/10.1016/j.bios.2015.07.014 View ArticleGoogle Scholar
- M.L. Wilson, L. Gaido, Laboratory diagnosis of urinary tract infections in adult patients. Clin. Infect. Dis. 38(8), 1150–1158 (2004). https://doi.org/10.1086/383029 View ArticleGoogle Scholar
- U. Jodal, U. Lindberg, K. Lincoln, Level diagnosis of symptomatic urinary tract infections in childhood. Acta Paediatr. 64(2), 201–208 (2008). https://doi.org/10.1111/j.1651-2227.1975.tb03822.x View ArticleGoogle Scholar
- S.P. Johnston, M.M. Ballard, M.J. Beach, L. Causer, P.P. Wilkins, Evaluation of three commercial assays for detection of Giardia and Cryptosporidium organisms in fecal specimens. J. Clin. Microbiol. 41(2), 623–626 (2003). https://doi.org/10.1128/JCM.41.2.623-626.2003 View ArticleGoogle Scholar
- A. Kinkhabwala, Z. Yu, S. Fan, Y. Avlasevich, K. Müllen, W.E. Moerner, Large single-molecule fluorescence enhancements produced by a bowtie nanoantenna. Nat. Photonics 3(11), 654–657 (2009). https://doi.org/10.1038/nphoton.2009.187 View ArticleGoogle Scholar
- H. Jin, D.A. Heller, M. Kalbacova, J.H. Kim, J. Zhang, A.A. Boghossian, N. Maheshri, M.S. Strano, Detection of single-molecule H2O2 signalling from epidermal growth factor receptor using fluorescent single-walled carbon nanotubes. Nat. Nanotechnol. 5, 302–309 (2010). https://doi.org/10.1038/nnano.2010.24 View ArticleGoogle Scholar
- L.P. McGuinness, Y. Yan, A. Stacey, D.A. Simpson, L.T. Hall, D. Maclaurin, S. Prawer, P. Mulvaney, J. Wrachtrup, F. Caruso, R.E. Scholten, L.C.L. Hollenberg, Quantum measurement and orientation tracking of fluorescent nanodiamonds inside living cells. Nat. Nanotechnol. 6(6), 358–363 (2011). https://doi.org/10.1038/nnano.2011.64 View ArticleGoogle Scholar
- S. Tabassum, W.M. Al-Asbahy, M. Afzal, F. Arjmand, R.H. Khan, Interaction and photo-induced cleavage studies of a copper based chemotherapeutic drug with human serum albumin: spectroscopic and molecular docking study. Mol. BioSyst. 8(9), 2424–2433 (2012). https://doi.org/10.1039/C2MB25119A View ArticleGoogle Scholar
- Z. Li, Y. Wang, J. Wang, Z. Tang, J.G. Pounds, Y. Lin, Rapid and sensitive detection of protein biomarker using a portable fluorescence biosensor based on quantum dots and a lateral flow test strip. Anal. Chem. 82(16), 7008–7014 (2010). https://doi.org/10.1021/ac101405a View ArticleGoogle Scholar
- J.A. Hansen, J. Wang, A.N. Kawde, Y. Xiang, K.V. Gothelf, G. Collins, Quantum-dot/aptamer-based ultrasensitive multi-analyte electrochemical biosensor. J. Am. Chem. Soc. 128(7), 2228–2229 (2006). https://doi.org/10.1021/ja060005h View ArticleGoogle Scholar
- A. Zajac, D. Song, W. Qian, T. Zhukov, Protein microarrays and quantum dot probes for early cancer detection. Colloids Surf. B 58(2), 309–314 (2007). https://doi.org/10.1016/j.colsurfb.2007.02.019 View ArticleGoogle Scholar
- M. Zhuang, C. Ding, A. Zhu, Y. Tian, Ratiometric fluorescence probe for monitoring hydroxyl radical in live cells based on gold nanoclusters. Anal. Chem. 86(3), 1829–1836 (2014). https://doi.org/10.1021/ac403810g View ArticleGoogle Scholar
- Y. Wang, L. Ge, P. Wang, M. Yan, S. Ge, N. Li, J. Yu, J. Huang, Photoelectrochemical lab-on-paper device equipped with a porous Au-paper electrode and fluidic delay-switch for sensitive detection of DNA hybridization. Lab Chip 13(19), 3945–3955 (2013). https://doi.org/10.1039/C3LC50430A View ArticleGoogle Scholar
- R.R. Anjana, J.S.A. Devi, M. Jayasree, R.S. Aparna, B. Aswathy, G.L. Praveen, G.M. Lekha, G. Sony, S, N-doped carbon dots as a fluorescent probe for bilirubin. Microchim. Acta 185, 11 (2018). https://doi.org/10.1007/s00604-017-2574-8 View ArticleGoogle Scholar
- M. Ferrari, Cancer nanotechnology: opportunities and challenges. Nat. Rev. Cancer 5(3), 161–171 (2005). https://doi.org/10.1038/nrc1566 View ArticleGoogle Scholar
- M. Wu, Q. Lai, Q. Ju, L. Li, H.D. Yu, W. Huang, Paper-based fluorogenic devices for in vitro diagnostics. Biosens. Bioelectron. 102, 256–266 (2018). https://doi.org/10.1016/j.bios.2017.11.006 View ArticleGoogle Scholar
- S. Fiorito, A. Serafino, F. Andreola, A. Togna, G. Togna, Toxicity and biocompatibility of carbon nanoparticles. J. Nanosci. Nanotechnol. 6(3), 591–599 (2006). https://doi.org/10.1166/jnn.2006.125 View ArticleGoogle Scholar
- S. Murugesan, T.J. Park, H. Yang, S. Mousa, R.J. Linhardt, Blood compatible carbon nanotubes—nano-based neoproteoglycans. Langmuir 22(8), 3461–3463 (2006). https://doi.org/10.1021/la0534468 View ArticleGoogle Scholar
- R. Shukla, V. Bansal, M. Chaudhary, A. Basu, R.R. Bhonde, M. Sastry, Biocompatibility of gold nanoparticles and their endocytotic fate inside the cellular compartment: a microscopic overview. Langmuir 21(3), 10644–10654 (2005). https://doi.org/10.1021/la0513712 View ArticleGoogle Scholar
- X. Hu, X. Gao, Multilayer coating of gold nanorods for combined stability and biocompatibility. Phys. Chem. Chem. Phys. 13(21), 10028–10035 (2011). https://doi.org/10.1039/C0CP02434A View ArticleGoogle Scholar
- A.W. Martinez, S.T. Phillips, E. Carrilho, S.W. Thomas, H. Sindi, G.M. Whitesides, Simple telemedicine for developing regions: camera phones and paper-based microfluidic devices for real-time, off-site diagnosis. Anal. Chem. 80(10), 3699–3707 (2008). https://doi.org/10.1021/ac800112r View ArticleGoogle Scholar
- A.K. Yetisen, J.L. Martinez-Hurtado, A. Garcia-Melendrez, F. da Cruz Vasconcellos, C.R. Lowe, A smartphone algorithm with inter-phone repeatability for the analysis of colorimetric tests. Sens. Actuators B-Chem. 196, 156–160 (2014). https://doi.org/10.1016/j.snb.2014.01.077 View ArticleGoogle Scholar
- X. Xu, A. Akay, H. Wei, S. Wang, B. Pingguan-Murphy, B.E. Erlandsson, X. Li, W. Lee, J. Hu, L. Wang, F. Xu, Advances in smartphone-based point-of-care diagnostics. Proc. IEEE 103(2), 236–247 (2015). https://doi.org/10.1109/JPROC.2014.2378776 View ArticleGoogle Scholar
- K.E. McCracken, J.Y. Yoon, Recent approaches for optical smartphone sensing in resource-limited settings: a brief review. Anal. Meth. 8(36), 6591–6601 (2016). https://doi.org/10.1039/C6AY01575A View ArticleGoogle Scholar
- S.K. Vashist, O. Mudanyali, E.M. Schneider, R. Zengerle, A. Ozcan, Cellphone-based devices for bioanalytical sciences. Anal. Bioanal. Chem. 406(14), 3263–3277 (2014). https://doi.org/10.1007/s00216-013-7473-1 View ArticleGoogle Scholar
- D. Paliy, A. Foi, R. Bilcu, V. Katkovnik, Denoising and interpolation of noisy Bayer data with adaptive cross-color filters. Proc. SPIE 6822, 68221K (2008). https://doi.org/10.1117/12.766217 View ArticleGoogle Scholar
- X. Jin, Z. Liu, J. Chen, CMOS vision sensor with fully digital image process integrated into low power 1/8-inch chip. Chin. Opt. Lett. 8(3), 282–285 (2010). https://doi.org/10.3788/COL20100803.0 View ArticleGoogle Scholar
- R. Fontaine, A survey of enabling technologies in successful consumer digital imaging products. In: Proceedings of the international image sensors workshop, Hiroshima, Japan, 30 May—2 June 2017 (2017)Google Scholar
- S.J. Qin, B. Yan, The point-of-care colorimetric detection of the biomarker of phenylamine in the human urine based on Tb3+ functionalized metal-organic framework. Anal. Chim. Acta 1012, 82–89 (2018). https://doi.org/10.1016/j.aca.2018.01.041 View ArticleGoogle Scholar
- H. Xu, K. Zhang, Q. Liu, Y. Liu, M. Xie, Visual and fluorescent detection of mercury ions by using a dually emissive ratiometric nanohybrid containing carbon dots and CdTe quantum dots. Microchim. Acta 184(4), 1199–1206 (2017). https://doi.org/10.1007/s00604-017-2099-1 View ArticleGoogle Scholar
- X. Wang, S. Wang, K. Huang, Z. Liu, Y. Gao, W. Zeng, A ratiometric upconversion nanosensor for visualized point-of-care assay of organophosphonate nerve agent. Sens. Actuators B-Chem. 241, 1188–1193 (2017). https://doi.org/10.1016/j.snb.2016.10.015 View ArticleGoogle Scholar
- P. Das, U.J. Krull, Detection of a cancer biomarker protein on modified cellulose paper by fluorescence using aptamer-linked quantum dots. Analyst 142(17), 3132–3135 (2017). https://doi.org/10.1039/c7an00624a View ArticleGoogle Scholar
- X. Weng, S. Neethirajan, Aptamer-based fluorometric determination of norovirus using a paper-based microfluidic device. Microchim. Acta 184(11), 4545–4552 (2017). https://doi.org/10.1007/s00604-017-2467-x View ArticleGoogle Scholar
- B. Li, X. Zhou, H. Liu, H. Deng, R. Huang, D. Xing, Simultaneous detection of antibiotic resistance genes on paper-based chip using [Ru(phen)2d ppz]2+ turn-on fluorescence probe. ACS Appl. Mater. Interfaces. 10(5), 4494–4501 (2018). https://doi.org/10.1021/acsami.7b17653 View ArticleGoogle Scholar
- Y. Seok, H.A. Joung, J.Y. Byun, H.S. Jeon, S.J. Shin, S. Kim, Y.B. Shin, H.S. Han, M.G. Kim, A paper-based device for performing loop-mediated isothermal amplification with real-time simultaneous detection of multiple DNA targets. Theranostics 7(8), 2220–2230 (2017). https://doi.org/10.7150/thno.18675 View ArticleGoogle Scholar
- K. Salama, H. Eltoukhy, A. Hassibi, A.E. Gamal, Modeling and simulation of luminescence detection platforms. Biosens. Bioelectron. 19(11), 1377–1386 (2004). https://doi.org/10.1016/j.bios.2003.12.031 View ArticleGoogle Scholar
- X.F. Li, Q.H. Wang, D.H. Li, A.H. Wang, Image processing to eliminate crosstalk between neighboring view images in three-dimensional lenticular display. J. Disp. Technol. 7(8), 443–447 (2011). https://doi.org/10.1109/JDT.2011.2142174 View ArticleGoogle Scholar
- L. Shen, J.A. Hagen, I. Papautsky, Point-of-care colorimetric detection with a smartphone. Lab Chip 12(21), 4240–4243 (2012). https://doi.org/10.1039/C2LC40741H View ArticleGoogle Scholar
- K.E. McCracken, S.V. Angus, K.A. Reynolds, J.Y. Yoon, Multimodal imaging and lighting bias correction for improved μPAD-based water quality monitoring via smartphones. Sci. Rep. 6, 27529 (2016). https://doi.org/10.1038/srep27529 View ArticleGoogle Scholar
- R.B. Sekar, A. Periasamy, Fluorescence resonance energy transfer (FRET) microscopy imaging of live cell protein localizations. J. Cell Biol. 160(5), 629–633 (2003). https://doi.org/10.1083/jcb.200210140 View ArticleGoogle Scholar
- B.A. Pollok, R. Heim, Using GFP in FRET-based applications. Trends Cell Biol. 9(2), 57–60 (1999). https://doi.org/10.1016/S0962-8924(98)01434-2 View ArticleGoogle Scholar
- E.A. Jares-Erijman, T.M. Jovin, FRET imaging. Nat. Biotechnol. 21(11), 1387–1395 (2003). https://doi.org/10.1038/nbt896 View ArticleGoogle Scholar
- M.O. Noor, U.J. Krull, Camera-based ratiometric fluorescence transduction of nucleic acid hybridization with reagentless signal amplification on a paper-based platform using immobilized quantum dots as donors. Anal. Chem. 86(20), 10331–10339 (2014). https://doi.org/10.1021/ac502677n View ArticleGoogle Scholar
- S.A. Díaz, L. Giordano, T.M. Jovin, E.A. Jares-Erijman, Modulation of a photoswitchable dual-color quantum dot containing a photochromic FRET acceptor and an internal standard. Nano Lett. 12(7), 3537–3544 (2012). https://doi.org/10.1021/nl301093s View ArticleGoogle Scholar
- Q.X. Wang, S.F. Xue, Z.H. Chen, S.H. Ma, S. Zhang, G. Shi, M. Zhang, Dual lanthanide-doped complexes: the development of a time-resolved ratiometric fluorescent probe for anthrax biomarker and a paper-based visual sensor. Biosens. Bioelectron. 94, 388–393 (2017). https://doi.org/10.1016/j.bios.2017.03.027 View ArticleGoogle Scholar
- C.M. Tyrakowski, P.T. Snee, Ratiometric CdSe/ZnS quantum dot protein sensor. Anal. Chem. 86(5), 2380–2386 (2014). https://doi.org/10.1021/ac4040357 View ArticleGoogle Scholar
- K. Wang, J. Qian, D. Jiang, Z. Yang, X. Du, K. Wang, Onsite naked eye determination of cysteine and homocysteine using quencher displacement-induced fluorescence recovery of the dual-emission hybrid probes with desired intensity ratio. Biosens. Bioelectron. 65, 83–90 (2015). https://doi.org/10.1016/j.bios.2014.09.093 View ArticleGoogle Scholar
- W.R. Algar, M. Massey, U.J. Krull, The application of quantum dots, gold nanoparticles and molecular switches to optical nucleic-acid diagnostics. Trends Anal. Chem. 28(3), 292–306 (2009). https://doi.org/10.1016/j.trac.2008.11.012 View ArticleGoogle Scholar
- X. Yu, L. Yang, T. Zhao, R. Zhang, L. Yang, C. Jiang, J. Zhao, B. Liu, Z. Zhang, Multicolorful ratiometric-fluorescent test paper for determination of fluoride ions in environmental water. RSC Adv. 7(84), 53379–53384 (2017). https://doi.org/10.1039/C7RA09972J View ArticleGoogle Scholar
- M. Dou, D.C. Dominguez, X. Li, J. Sanchez, G. Scott, A versatile PDMS/paper hybrid microfluidic platform for sensitive infectious disease diagnosis. Anal. Chem. 86(15), 7978–7986 (2014). https://doi.org/10.1021/ac5021694 View ArticleGoogle Scholar
- M.G. Caglayan, S. Sheykhi, L. Mosca, P. Anzenbacher, Fluorescent zinc and copper complexes for detection of adrafinil in paper-based microfluidic devices. Chem. Commun. 52(53), 8279–8282 (2016). https://doi.org/10.1039/C6CC03640F View ArticleGoogle Scholar
- S.J. Yeo, K. Choi, B.T. Cuc, N.N. Hong, D.T. Bao, N.M. Ngoc, M.Q. Le, N.L.K. Hang, N.C. Thach, S.K. Mallik, H.S. Kim, C.K. Chong, H.S. Choi, H.W. Sung, K. Yu, H. Park, Smartphone-based fluorescent diagnostic system for highly pathogenic H5N1 viruses. Theranostics 6(2), 231–242 (2016). https://doi.org/10.7150/thno.14023 View ArticleGoogle Scholar
- H.C. Koydemir, Z. Gorocs, D. Tseng, B. Cortazar, S. Feng, R.Y.L. Chan, J. Burbano, E. McLeod, A. Ozcan, Rapid imaging, detection and quantification of Giardia lamblia cysts using mobile-phone based fluorescent microscopy and machine learning. Lab Chip 15(5), 1284–1293 (2015). https://doi.org/10.1039/C4LC01358A View ArticleGoogle Scholar
- A. Hossain, J. Canning, S. Ast, P.J. Rutledge, T.L. Yen, A. Jamalipour, Lab-in-a-phone: smartphone-based portable fluorometer for pH measurements of environmental water. IEEE Sens. J. 15(9), 5095–5102 (2015). https://doi.org/10.1109/JSEN.2014.2361651 View ArticleGoogle Scholar
- Y. Koo, J. Sankar, Y. Yun, High performance magnesium anode in paper-based microfluidic battery, powering on-chip fluorescence assay. Biomicrofluidics 8(5), 054104 (2014). https://doi.org/10.1063/1.4894784 View ArticleGoogle Scholar
- N.K. Thom, K. Yeung, M.B. Pillion, S.T. Phillips, “Fluidic batteries” as low-cost sources of power in paper-based microfluidic devices. Lab Chip 12(10), 1768–1770 (2012). https://doi.org/10.1039/C2LC40126F View ArticleGoogle Scholar
- N.K. Thom, G.G. Lewis, K. Yeung, S.T. Phillips, Quantitative fluorescence assays using a self-powered paper-based microfluidic device and a camera-equipped cellular phone. RSC Adv. 4(3), 1334–1340 (2014). https://doi.org/10.1039/C3RA44717K View ArticleGoogle Scholar
- K.E. McCracken, T. Tat, V. Paz, J.Y. Yoon, Smartphone-based fluorescence detection of bisphenol A from water samples. RSC Adv. 7(15), 9237–9243 (2017). https://doi.org/10.1039/C6RA27726H View ArticleGoogle Scholar
- K.D. Long, E.V. Woodburn, H.M. Le, U.K. Shah, S.S. Lumetta, B.T. Cunningham, Multimode smartphone biosensing: the transmission, reflection, and intensity spectral (TRI)-analyzer. Lab Chip 17(19), 3246–3257 (2017). https://doi.org/10.1039/c7lc00633k View ArticleGoogle Scholar
- E. Petryayeva, W.R. Algar, A job for quantum dots: use of a smartphone and 3D-printed accessory for all-in-one excitation and imaging of photoluminescence. Anal. Bioanal. Chem. 408(11), 2913–2925 (2016). https://doi.org/10.1007/s00216-015-9300-3 View ArticleGoogle Scholar
- M.A. Hossain, J. Canning, S. Ast, K. Cook, P.J. Rutledge, A. Jamalipour, Combined “dual” absorption and fluorescence smartphone spectrometers. Opt. Lett. 40(8), 1737–1740 (2015). https://doi.org/10.1364/OL.40.001737 View ArticleGoogle Scholar
- J. Canning, A. Lau, M. Naqshbandi, I. Petermann, M.J. Crossley, Measurement of fluorescence in a rhodamine-123 doped self-assembled “giant” mesostructured silica sphere using a smartphone as optical hardware. Sensors 11(7), 7055–7062 (2011). https://doi.org/10.3390/s110707055 View ArticleGoogle Scholar
- Y. Dattner, O. Yadid-Pecht, Low light CMOS contact imager with an integrated poly-acrylic emission filter for fluorescence detection. Sensors 10(5), 5014–5027 (2010). https://doi.org/10.3390/s100505014 View ArticleGoogle Scholar
- C. Richard, A. Renaudin, V. Aimez, P.G. Charette, An integrated hybrid interference and absorption filter for fluorescence detection in lab-on-a-chip devices. Lab Chip 9(10), 1371–1376 (2009). https://doi.org/10.1039/B819080A View ArticleGoogle Scholar
- M.L. Adams, M. Enzelberger, S. Quake, A. Scherer, Microfluidic integration on detector arrays for absorption and fluorescence micro-spectrometers. Sens. Actuators A-Phys. 104, 25–31 (2003). https://doi.org/10.1016/S0924-4247(02)00477-6 View ArticleGoogle Scholar
- O. Hofmann, X. Wang, A. Cornwell, S. Beecher, A. Raja, D.D.C. Bradley, A.J. de Mello, J.C. de Mello, Monolithically integrated dye-doped PDMS long-pass filters for disposable on-chip fluorescence detection. Lab Chip 6(8), 981–987 (2006). https://doi.org/10.1039/B603678C View ArticleGoogle Scholar
- M. Beiderman, T. Tam, A. Fish, G.A. Jullien, O. Yadid-Pecht, A low-light CMOS contact imager with an emission filter for biosensing applications. IEEE Trans. Biomed. Circuits Syst. 2(3), 193–203 (2008). https://doi.org/10.1109/TBCAS.2008.2001866 View ArticleGoogle Scholar
- S. Pang, C. Han, L.M. Lee, C. Yang, Fluorescence microscopy imaging with a Fresnel zone plate array based optofluidic microscope. Lab Chip 11(21), 3698–3702 (2011). https://doi.org/10.1039/C1LC20654K View ArticleGoogle Scholar
- S.A. Lee, X. Ou, J.E. Lee, C. Yang, Chip-scale fluorescence microscope based on a silo-filter complementary metal-oxide semiconductor image sensor. Opt. Lett. 38(11), 1817–1819 (2013). https://doi.org/10.1364/OL.38.001817 View ArticleGoogle Scholar
- Y.J. Hung II, C.C.Davis Smolyaninov, H.C. Wu, Fluorescence enhancement by surface gratings. Opt. Express 14(22), 10825–10830 (2006). https://doi.org/10.1364/OE.14.010825 View ArticleGoogle Scholar
- D. Gallegos, K.D. Long, H. Yu, P.P. Clark, Y. Lin, S. George, P. Nath, B.T. Cunningham, Label-free biodetection using a smartphone. Lab Chip 13(11), 2124–2132 (2013). https://doi.org/10.1039/C3LC40991K View ArticleGoogle Scholar
- S. Ricciardi, F. Frascella, A. Angelini, A. Lamberti, P. Munzert, L. Boarino, R. Rizzo, A. Tommasi, E. Descrovi, Optofluidic chip for surface wave-based fluorescence sensing. Sens. Actuators B Chem. 215, 225–230 (2015). https://doi.org/10.1016/j.snb.2015.03.063 View ArticleGoogle Scholar
- B.R. Schudel, C.J. Choi, B.T. Cunningham, P.J.A. Kenis, Microfluidic chip for combinatorial mixing and screening of assays. Lab Chip 9(12), 1676–1680 (2009). https://doi.org/10.1039/B901999E View ArticleGoogle Scholar
- T.L. Danielson, G.D. Rayson, D.M. Anderson, R. Estell, E.L. Fredrickson, B.S. Green, Impact of filter paper on fluorescence measurements of buffered saline filtrates. Talanta 59(3), 601–604 (2003). https://doi.org/10.1016/S0039-9140(02)00575-1 View ArticleGoogle Scholar
- N. Guo, K.W. Cheung, H.T. Wong, D. Ho, CMOS time-resolved, contact, and multispectral fluorescence imaging for DNA molecular diagnostics. Sensors 14(11), 20602–20619 (2014). https://doi.org/10.3390/s141120602 View ArticleGoogle Scholar
- S. Bouccara, A. Fragola, E. Giovanelli, G. Sitbon, N. Lequeux, T. Pons, V. Loriette, Time-gated cell imaging using long lifetime near-infrared-emitting quantum dots for autofluorescence rejection. J. Biomed. Opt. 19(5), 051208 (2014). https://doi.org/10.1117/1.JBO.19.5.051208 View ArticleGoogle Scholar
- Q. Ju, Y. Liu, D. Tu, H. Zhu, R. Li, X. Chen, Lanthanide-doped multicolor GdF3 nanocrystals for time-resolved photoluminescent biodetection. Chem. Eur. J. 17(31), 8549–8554 (2011). https://doi.org/10.1002/chem.201101170 View ArticleGoogle Scholar
- H. Kim, E. Petryayeva, W.R. Algar, Enhancement of quantum dot Forster resonance energy transfer within paper matrices and application to proteolytic assays. IEEE J. Sel. Top. Quantum Electron. 20(3), 141–151 (2014). https://doi.org/10.1109/JSTQE.2013.2280498 View ArticleGoogle Scholar
- A.S. Paterson, B. Raja, V. Mandadi, B. Townsend, M. Lee, A. Buell, B. Vu, J. Brgoch, R.C. Willson, A low-cost smartphone-based platform for highly sensitive point-of-care testing with persistent luminescent phosphors. Lab Chip 17(6), 1051–1059 (2017). https://doi.org/10.1039/c6lc01167e View ArticleGoogle Scholar
- K.G. Shah, P. Yager, Wavelengths and lifetimes of paper autofluorescence: a simple substrate screening process to enhance the sensitivity of fluorescence-based assays in paper. Anal. Chem. 89(22), 12023–12029 (2017). https://doi.org/10.1021/acs.analchem.7b02424 View ArticleGoogle Scholar
- F. Zhou, M.O. Noor, U.J. Krull, Luminescence resonance energy transfer-based nucleic acid hybridization assay on cellulose paper with upconverting phosphor as donors. Anal. Chem. 86(5), 2719–2726 (2014). https://doi.org/10.1021/ac404129t View ArticleGoogle Scholar
- C.R. Ispas, G. Crivat, S. Andreescu, Review: recent developments in enzyme-based biosensors for biomedical analysis. Anal. Lett. 45(2), 168–186 (2012). https://doi.org/10.1080/00032719.2011.633188 View ArticleGoogle Scholar
- M. He, Z. Liu, Paper-based microfluidic device with upconversion fluorescence assay. Anal. Chem. 85(24), 11691–11694 (2013). https://doi.org/10.1021/ac403693g View ArticleGoogle Scholar
- L. Wang, R. Yan, Z. Huo, L. Wang, J. Zeng, J. Bao, X. Wang, Q. Peng, Y. Li, Fluorescence resonant energy transfer biosensor based on upconversion-luminescent nanoparticles. Angew. Chem. Int. Ed. 44(37), 6054–6057 (2005). https://doi.org/10.1002/anie.200501907 View ArticleGoogle Scholar
- W.W. Yu, I.M. White, Inkjet-printed paper-based SERS dipsticks and swabs for trace chemical detection. Analyst 138(4), 1020–1025 (2013). https://doi.org/10.1039/C2AN36116G View ArticleGoogle Scholar
- J. Qiang, U. Uvaraj, K. Ulrich, Paper-based DNA detection using lanthanide-doped LiYF4 upconversion nanocrystals as bioprobe. Small 10(19), 3912–3917 (2014). https://doi.org/10.1002/smll.201400683 View ArticleGoogle Scholar
- S. Doughan, U. Uddayasankar, U.J. Krull, A paper-based resonance energy transfer nucleic acid hybridization assay using upconversion nanoparticles as donors and quantum dots as acceptors. Anal. Chim. Acta 878, 1–8 (2015). https://doi.org/10.1016/j.aca.2015.04.036 View ArticleGoogle Scholar
- J.P. Golden, F.S. Ligler, A comparison of imaging methods for use in an array biosensor. Biosens. Bioelectron. 17(9), 719–725 (2002). https://doi.org/10.1016/S0956-5663(02)00060-X View ArticleGoogle Scholar
- B. Jang, P. Cao, A. Chevalier, A. Ellington, A. Hassibi, A CMOS fluorescent-based biosensor microarray. in 2009 IEEE international solid-state circuits conference (2009), pp. 436–437. https://doi.org/10.1109/isscc.2009.4977495
- G. Giraud, H. Schulze, D.U. Li, T.T. Bachmann, J. Crain, D. Tyndall, J. Richardson, R. Walker, D. Stoppa, E. Charbon, R. Henderson, J. Arlt, Fluorescence lifetime biosensing with DNA microarrays and a CMOS-SPAD imager. Biomed. Opt. Express 1(5), 1302–1308 (2010). https://doi.org/10.1364/BOE.1.001302 View ArticleGoogle Scholar
- A.E. Cetin, A.F. Coskun, B.C. Galarreta, M. Huang, D. Herman, A. Ozcan, H. Altug, Handheld high-throughput plasmonic biosensor using computational on-chip imaging. Light Sci. Appl. 3, e122 (2014). https://doi.org/10.1038/lsa.2014.3 View ArticleGoogle Scholar
- M.W. Seo, K. Kagawa, K. Yasutomi, Y. Kawata, N. Teranishi, Z. Li, I.A. Halin, S. Kawahito, A 10 ps time-resolution CMOS image sensor with two-tap true-CDS lock-in pixels for fluorescence lifetime imaging. IEEE J. Solid-State Circuits 51, 141–154 (2016). https://doi.org/10.1109/JSSC.2015.2496788 View ArticleGoogle Scholar
- H. Takehara, O. Kazutaka, M. Haruta, T. Noda, K. Sasagawa, T. Tokuda, J. Ohta, On-chip cell analysis platform: implementation of contact fluorescence microscopy in microfluidic chips. AIP Adv. 7(9), 095213 (2017). https://doi.org/10.1063/1.4986872 View ArticleGoogle Scholar
- W. Li, T. Knoll, A. Sossalla, H. Bueth, H. Thielecke, On-chip integrated lensless fluorescence microscopy/spectroscopy module for cell-based sensors. Proc. SPIE 7894, 78940Q (2011). https://doi.org/10.1117/12.875417 View ArticleGoogle Scholar
- S.V. Kesavan, C.P. Allier, F. Navarro, F. Mittler, B. Chalmond, J.M. Dinten, Lensless imaging system to quantify cell proliferation. Proc. SPIE 8587, 858708 (2013). https://doi.org/10.1117/12.2001826 View ArticleGoogle Scholar
- G. Zheng, S.A. Lee, Y. Antebi, M.B. Elowitz, C. Yang, The ePetri dish, an on-chip cell imaging platform based on subpixel perspective sweeping microscopy (SPSM). Proc. Natl. Acad. Sci. USA 108(41), 16889–16894 (2011). https://doi.org/10.1073/pnas.1110681108 View ArticleGoogle Scholar
- A. Jain, O. Taghavian, D. Vallejo, E. Dotsey, D. Schwartz, F.G. Bell, C. Greef, D.H. Davies, J. Grudzien, A.P. Lee, P. Felgner, L. Liang, Evaluation of quantum dot immunofluorescence and a digital CMOS imaging system as an alternative to conventional organic fluorescence dyes and laser scanning for quantifying protein microarrays. J. Proteom. 16(8), 1271–1279 (2016). https://doi.org/10.1002/pmic.201500375 View ArticleGoogle Scholar